Don't let the tiny size fool you because this is a fully featured mixed signal test & measurement system.
It's a complete USB oscilloscope, logic analyzer, arbitrary waveform generator and spectrum analyzer all rolled into one !
We call it BS10. It's a problem solver that fits in the palm of your hand.
BS10 provides both 5V & 3.3V, I/O & clock signals so it can power, control and even clock logic or microcontroller circuits!
It has 100 MHz bandwidth, up to 40 MSps logic, simultaneous waveform and clock generation and offers serial control of connected devices.
It's the ideal tool for prototyping analog electronics projects, digital and logic circuits or Arduino, Raspberry Pi and other microcontroller based systems with busses, sensors, servos and I/O.
All you need to prototype new designs are basic electronic components, breadboards and wires and sometimes, a Raspberry Pi or Arduino board and some accessories. If you have a spare Raspberry Pi you may not even need a PC!
What you won't need are oscilloscopes, logic analyzers, waveform generators, power supplies or spectrum analyzers because that's what BitScope provides via it's Smart Port Interface.
BS10 is very fast with a frame rate up to 100 Hz driving a digital phosphor display. It works like a quality stand-alone mixed signal scope. View waveforms, plots, spectra and more on its smooth flowing real-time display. Even live captured logic data can be viewed this way.
With its large buffers it can support very high speed one-shot capture for such a small device with post-capture zoom, scrolling and measurement. Alternatively stream capture direct to disk for off-line replay and analysis.
BitScope DSO and Logic software applications are included as standard. They are compatible with Windows, Mac OS X, Linux and Raspberry Pi and a growing range of other applications are also available for download at no extra cost.
BitScope Mini may be small but it packs in a huge range of features...
BS10 is far more than just a logic protocol analyzer; it's a fully featured portable oscilloscope, spectrum analyzer, waveform generator, power supply and logic analyzer in a compact USB powered unit. It has multi-range analog channels with adjustable input offsets, level adjustable logic comparators for two of its eight logic channels and it's compatible with the dual channel active differential probe making possible differential measurements not otherwise possible, especially in such a compact instrument.
BitScope Mini merges mixed signal capture and generation with protocol analysis.
At its heart is the BitScope DSP Capture Engine. This is responsible for the primary waveform and protocol capture functions and synchronizes with the mixed signal trigger and waveform and clock generators. It also manages the logic and analog buffers, input channel ranges and offsets.
Working in concert with the BitScope software applications on the host computer, BS10's signal acquisition, processing and sythesis components operate together to implement a wide range of test and measurement instruments many of which can operate at the same time.
One of the simplest and probably most useful test instruments. This example shows an 8 kHz sine and triangle waveform, sythesized by BS10's waveform generator and sampled at 20 MHz via its two analog inputs using BitScope DSO.
Like a regular oscilloscope, BS10 and DSO provide familiar trigger, timebase and voltage scaling and offset controls with display refresh rates up to 100Hz.
Of course BS10 is a BitScope so it also has 8 logic channels in addition to the analog channels. Below an 1 kHz sinusoid generated by a microcontroller in 32 steps is shown with its analog output via an 8 bit D/A. The logic channels show the D/A input (digital signals) and the analog channels show the op-amp buffered D/A output and its inverse.
This mixed signal display is another virtual instrument of BitScope DSO. However if your primary focus is protocol analysis you may prefer to use BitScope Logic.
When looking at serial logic protocols such as SPI, I2C, CAN or asynchronous serial such as RS-232 or 422 etc, BitScope Logic is a good choice.
This app is similar to DSO's mixed signal instrument but offers full protocol packet decoding and display. BitScope Logic and BS10 can display decoded logic on an ongoing basis live as it's captured, frame by frame (unlike any other pocket sized analyzer we know of).
BitScope Mini has a massive analog bandwidth for such a small product and this can be used to good effect via the sub-sampling modes of BitScope DSO. Sub-sampling is a powerful analysis technique for periodic waveforms. It is supported by BitScope Mini because its A/D is extremely fast (3ns aperture). Signals up to and even beyond 100 MHz can be displayed with equivalent time sample rates beyond 2GHz so long as the signal is periodic and has a frequency, amplitude and harmonic structure which does not change rapidly for the duration of the sub-sampled capture frame.
Sub-sampling can be particularly useful when the signal does change slowly during the capture frame. For example, this display shows a single sub-sampled turn-on period of a 20 MHz crystal oscillator sub-sampled with an effective sample rate of 1GHz. This waveform is not what you'd see if you used a 1GHz real-time sampling oscilloscope.
Arguably sub-sampling provides more useful information in this case; you see a frequency shifted version of the waveform over the entire duration of the clock turn-on. Each cycle is representative of hundreds or thousands of almost identical cycles at the original much higher frequency.
It's important to understand that sub-sampling is not a replacement for real-time capture but in some situations like this one it's useful. To see this turn-on envelope using a real-time sampling oscilloscope would require a massive capture buffer (to record all the data at a 1GHz sample rate). Only the most expensive real-time storage scopes with huge capture buffers can do this and real-time capture is unlikely to tell you more than simply sub-sampling the quasi-period clock turn-on signal in the first place.
BS10 inherits BitGen Waveform Generation features of BS120.
Like BitGen it generates arbitrary analog waveforms and clocks signals simultaneously. This example shows a 16 step sinusoid and the clock that generates it captured on the analog channels. Up to 1024 steps (samples) per period are supported with arbitrary (user programmable) data.
BS10 can be used as a powerful real-time mixed domain (time and frequency) spectrum analyzer. It can be used in baseband or narrow band (RF) applications.
The example above shows white noise passed through a 20 kHz audio amplifier where the roll off starts at 15 kHz to -3dB at 17 kHz, -20dB at 20kHz and -50dB beyond 22kHz.
The examples above are just some of the test and measurement possibilities available because despite its tiny size, BS10 is a fully featured BitScope!
For current and future software options for BS10 please see our software pages.
|Build a powerful mixed signal workstation with one simple 3D print !||Sep 3|
|Raspberry Pi Zero W, a wireless wonder for only $10 !||Feb 28|
|BitScope, Blade & Raspberry Pi at the Sydney Maker Faire!||Aug 20|
|Build servers with Raspberry Pi and BitScope Blade.||Mar 16|
|Raspberry Pi 3 Model B unboxed & reviewed with Duo Pi.||Mar 07|
|Raspberry Pi 3 Launch and BitScope Blade||Feb 29|
|Physical Computing with Raspberry Pi and BitScope!||Dec 01|
|BitScope Micro with Raspberry Pi Touchscreen Display!||Sep 09|
|BitScope, Ozberry & Pimoroni at Sydney Mini Maker Faire!||Aug 12|
|BitScope Blade for Raspberry Pi at Hannover Maker Faire!||Jun 05|
|BitScope DSO 2.9 Now Available.||Jun 02|
|Getting Started with BitScope||May 08|
|BitScope ARM Platform Compatibility||Feb 10|
|Bootstrapping Raspberry Pi 2 for BitScope||Feb 05|
|Raspberry Pi 2 and BitScope Performance||Feb 02|
|BitScope Micro & Raspberry Pi in Education at BETT||Jan 21|
|Raspberry Pi Model A+ or B+ or B+ Slimline?||Nov 14|
|Electronic Measurement with BitScope & Raspberry Pi||Nov 12|
|Hands on with BitScope Micro at electronica 2014||Nov 11|
|BitScope Micro Video Tutorial with Farnell element14||Nov 10|
|BitScope Micro and Raspberry Pi at electronica 2014!||Nov 06|
|BitScope Micro element14 Webinar Review.||Nov 03|
|BitScope Micro Webinar with element14!||Oct 29|
|element14 launches BitScope Micro for Raspberry Pi!||Oct 27|
|BitScope Micro Diagnostic Port Explained.||Aug 16|
|MagPi talks Raspberry Pi B+, HATs & BitScope.||Aug 04|
|Raspberry Pi B+ BitScope x 4 = 8 analog + 32 logic !||Aug 01|
|BitScope Model 10 or BitScope Micro?||Apr 25|
|BitScope Micro, what's in the box?||Apr 17|
|BitScope Micro Questions & Answers||Apr 12|
|BitScope Micro (New Product) for Raspberry Pi!||Apr 11|
|BitScope Workshop at OzBerryPi||Apr 8|
|BitScope Library V2 API Guide Published||Mar 21|
|Optimizing real-time waveform display for Raspberry Pi||Mar 17|
|Volume purchase discounts now available online!||Mar 11|
|BitScope Documentation Updated Online||Mar 05|
|BitScope DSO Version 2.6 Production Release||Mar 03|
|Low cost networked data acquisition with Raspberry Pi.||Dec 17|
|BitScope Server, shared remote access for any BitScope!||Dec 15|
|BitScope DSO updated for Server Compatibility||Dec 14|
|Collaborative Solutions, Online & Interactive Education.||Dec 09|
|Powerhouse ThinkSpace Digital Learning||Dec 04|
|The Sydney Mini Maker Faire Review.||Nov 28|
|BitScope Ed at the Sydney Mini Maker Faire!||Nov 20|
|BitScope Library V2.0 and Proto Scope Application||Nov 15|
|BitScope DSO, Logic, Chart, Meter and Library updated||Nov 12|
|Automated Waveform Analysis with Raspberry Pi||Nov 09|
|Reducing Quantization Noise with BitScope||Nov 08|
|Breadboard One, the 4 Bit Up/Down Counter||Nov 07|
|BitScope Software upgraded for Mac OS X Mavericks||Nov 06|
|Using a USB BitScope with any Linux system.||Nov 05|
|BitScope DSO upgraded for Mac OS X Mavericks.||Nov 03|
|Sydney BitScope, Online Demo Updated.||Oct 30|
|BitScope Touch Screen User Interface||Oct 28|
|Network access for USB BitScopes via Raspberry Pi.||Oct 24|
|BitScope is virtually indestructible, just as we intended!||Oct 23|
|BitScope Ed at OzBerryPi, Presentation Review.||Oct 21|
|BitScope and friends at OzBerryPi Maker Space||Oct 20|
|Vertical Offsets and Dual Channel Display||Oct 19|
|How to Install BitScope DSO on Raspberry Pi||Oct 18|
|Breadboard One, a typical Mixed Signal Circuit.||Oct 17|
|Electronic Projects Lab for Raspberry Pi||Oct 17|
|Dual Channel Active Differential Probe (New Product).||Oct 15|
|BitScope Smart Port. The clever connection.||Oct 12|
|BitScope Ed, an Educational Blog about Electronics.||Oct 09|
|Why the oscilloscope update rate is important.||Oct 06|
|Visualizing sound with Sonic Pi and BitScope||Oct 05|
|BitScope Software Blog Launched||Oct 04|
|Differential measurements matter, here's why...||Oct 01|
|How to Install Raspbian and BitScope on Raspberry Pi||Sep 29|
|High speed data acquisition with Raspberry Pi||Sep 25|
|BitScope Software for Raspberry Pi Oscilloscope||Sep 27|
|BitScope DSO 2.7 update (for Raspberry Pi only)||Sep 24|
|BitScope Raspberry Pi Oscilloscope||Sep 20|